◆ Deep hemostasis, foot control outputs, mild hemostasis.
◆ According to the depth of hemostasis, the output mode stops automatically. The output power will not reduce when operating in dry and bleeding spot.
◆ Foot control mode: When the tissue reach the coagulation effects, the output power declines slowly.
◆ Auto mode: when the tissue or blood vessel was coagulated, output power drops off sharply, to keep the forceps from sticking and over burnt for tissues.
◆ Low voltage effectively reduces forming of the sparking of forceps, stimulating for the neuromuscular, to increase the security.
◆ Microcontroller system allows this unit can be used in many delicate surgery.
◆ Bipolar cut :macro bipolar.
Two working modes: bipolar coag、 bipolar cut
Power: 220V±22V(110V±11V) 50Hz±1Hz
Output power:1) bipolar coag:1-120W(load 200Ω)
2) bipolar cut:1-120W(load 200Ω)
None sticky bipolar Forceps and cable: 1set