It could clean and sterilize different high and low flux dialyzer, and inspect blood capacity automatically as wall as membrane broken experiment and pressure inspection, it can clean, sterilize and inspect two dialyzers at the same lime. With the printer function, it can print the inspected data out and stick on cleaned dialyzer to judge if the dialyzer can continue to use or not.
Switch on self-test function
Information memory of all aspect of dialyzer which facilitate to operate
Entirely new touch screen
Spare pump counter-rinse, to reach the best effect
Automatically inspect the precision of blood capacity sensor
Automatically inspect the precision of pressure sensor
Automatically inspect if exterior connector link lightly or not
Be capable to handle any kinds of hollow fiber dialyzer
(HDF)High efficiency and high-flux of hemodiafiltration dialyzer
Design for saving water and low inflow pressure to unload center water supply
Be provided with data saving when power cut
Professional maintenance interface which can inspect if all valves and pumps work normally or not Be provided with original creation function
With a special detector to inspect disinfection solution concentration, which assure the required disinfection
concentration filled with dialyzer to aviod the disinfection failure caused by manmade which occur medical accident
Be provided with alarm for block or fold of tube which assure liquid filling with dialyzer during disinfecting. With a highly security and complete clean
Contacting disinfection liquid automatically by system as well as rinsing and filling up dialyzer.
Main technical parameter
Exterior size: 480mm X 430mm X 560mm
Weight: About 25kgs
Circumstance temperature: 5~40°C
Relative humidity: 10%~80%
Pressure test: put 290mmHg±10mmHg pressure to one side of blood chamber.
Pressure release of dialyzer should be less than:
Low flux dialyzer: 0.83mmhg/sec
High flux dialyzer: 1.25mmhg/sec
Capacity test: Scope: 0~150ml, error: ±3%
Water required: Water temperature: 10°C~30°C
Water quality required: RO water reach the tandard5.1.2 of<
Water input pressure: 0.15MPa~0.35MPa
Water input flux: >1600ml/min
Rinsing time: 2min/time
Disinfection time: 7min/time
Reprocessing dialyzer time: 11 min/time
Water amount: about 8L/dialyzer
Power supply: AC220V
Disinfection solution: common disinfectant, ie， peracetic acid renalin and other special disinfectant.
Proposed use: 12%-15% peracetic acid or 100% renalin(alternative)
English: 6*8 lattice print rate: 1sec./row
Chinese: 12* 12 lattice.