What is the features of our dental autoclave?
1.Standard Class B with three times vacuum and drying, the remaining temperature of sterilized instrument is less than 0.2%.
2.The vacuum data can reach -0.8bar,it fits various instrument,including package,unpackaged,solid,hollow,multi-apertures and inset pipes. It makes sure the instrument can be sterilized completely.
3.Fuzzily computer controlled, digital bright show. The interface is modularized panel for easy operation.
4.It is installed with BOWIE&DICK which measures the penetration of water steam.
5.it has vacuum text which can make sure the ability of texting vacuuming.
6.Sterilization cycle self-comment.
7.Fast and independent steam generator.
8.Large capacity instrument tray made of SS. SU304.
9.Safety lock and triple temperature.overload protecting
10.Auto/Manual water adding
11.Inset cleaning device
12.Special function of creating medicine cotton and sterilizing rubber items
13.Handsome aspect and good quality
14.Cristal LCD display
15.5-Button panel with the LCD display, so convenient.
16.Warning message, Error is made easier to judge.
1.Please place the sterilization on a plane place , which could adjust the height by adjusting foot pad
2.Please adopt distilled water to extend the service life
3.Don't block the heat removal system on the surface of sterilization
4.Please put the instruments on the tray not touching one another allowing space for steam to pass through
5.Please empty the used water tank frequently. Usually it need empty when the clean water in the reservoir is empty.
6.Please ensure the door handle is closed tightly
7.Never try to open the chamber door if and when the pressure display doesn't show "0.0"
8.Please cut off the power and cool down the auto clave before maintenance
9.Don't drag the machine while moving
10.Power connection must be grounded properly
11.Must be provided enough power. (23L:2300VA/18L:1800VA)
12.While the outside temperature is too low, please pre-heating the machine without instruments for 30 minutes before sterilize
13.Don't cut off the power to stop the machine directly. If you need stop the machine directly, please press the "START/STOP" button for 3 seconds.
What is the specification of our class B dental autoclave?
Model (LCD/ LED)
220V/50Hz; 220V160Hz; 115V/60Hz;
Helix/BD Test; Vacuum Test;
Pre-Heating Time(From cold)
Approx 20 mins
Net /Gross Weight
Temperature: 5ºC-40ºC Relative Humidity: max 80%, Non condensing Height: max 3000 m (a.s.l.)
Capacity of the Distilled Water Reservoir
Approx. 5.0 L (water at level MAX)