Author：MeCan Medical–ultrasound machine manufacturers
1. The instrument cannot be turned off frequently. After shutting down for 30 seconds, it must be restarted after waiting for 30 seconds.
2. If the coagulation analyzer is no longer used for a long time, the instrument is packaged with the original packaging box, stored in the temperature -20-55 C, relative humidity
3. After booting, preheat for more than 30 minutes, so that the detection platform and preheating table can meet the temperature requirements of the instrument.
4. When using and changing the detection environment or reagent for the first time, the standard is fixed according to the reagent manual before the test.
5. Observe the AD value status before the test. If you are saturated, please do not adjust your own. You need to contact after -sales engineers for processing.
6. Do not open the test warehouse during the test.
7. After shutting down, the door of the testing table and the preheating platform should be closed.
8. Use towels to wet the surface and pallet guide rail gently with a towel in the diluted disinfection water daily.
9. Strictly prevent the washing liquid or other liquid into the instrument.
If you do nt pay attention to maintenance, this will bring a very heavy burden on the instrument, and it will shorten the service life and increase the cost of hospitals. Therefore, whether it is a coagulation analyzer or other medical equipment, it should be operated and maintained in accordance with regular procedures, so that the instrument can work better for the hospital.