Tel: +86 20 8483 5259
Whatsapp: +86 159 8923 0468
Medical Philippines Expo 2018
MEDIC EAST AFRICA 2018
What is the detail of our infant radiant warmer?
Three control modes: Pre-warm mode, manual mode
and baby mode controlled by micro-computer;
Set temperature and baby temperature can be displayed separately;
Tri-safety over-temp protection:
Double CPU control system.
Isolated over heating cutoff device.
Horizon angle of heater head and inclination of the bassinet can be adjusted;
The glass panels around bassinet can be disassemble or turned outward;
X-ray cassette under the bassinet;
With temperature data storage function
Neonate phototherapy equipment
Optional configuration: Vertical height adjustment stand
Mode of control: Pre-warm mode control, manual mode control and baby mode controlled by micro-computer.
Baby mode temperature control range:34.5~ 37.5℃
Temperature sensor display range: 5℃~65℃
Accuracy of skin temperature sensor: ≤0.3℃
Deviation between the measure by the skin temperature sensor
and control temperature: ≤0.5℃
Temperature uniformity of mattress: ≤2℃
Angle of warmer module: 0°,30°,60°for two ways
Inclination of the bassinet: ±10°
APGAR timer: it will give sound when the unit tones at 1’,5’,10’
Failure alarm: over temperature alarm, deviation alarm, sensor failure alarm, power failure alarm, setting alarm, checking alarm and so on.
Max.Bilirubin uniformity of radiating on the mattress effective range: >=1000uW/cm2
Bilirubin niformity of radiating on the mattress effective range:>0.4
Wave length of blue light: 420-470nm.
More infant radiant warmers for your reference