1.This Anesthesia Machine is an essential important anesthesia device in an operation room. Its function is to provide oxygen and anesthetic agent to a patient needing to go through an anesthesia operation and perform ventilation management.
This anesthesia machine is intended for use in children and adults with a body weight heavier than 10kg.
This Anesthesia Machine is equipped with an accurate dedicated anaesthetic vaporizer and a safety device for preventing cyanosis and necessary alarm system. During the anesthesia, the patient’s respiratory functions can be managed by using a microcomputer controlled pneumatic electrically controlled synchronizing anesthesia respirator. Each connection part of the whole machine is a standard interface. A highly efficient and large volume soda lime absorber can minimize the re-inhalation of carbon dioxide by the patient.
Operating conditions for JINLING-01 I Anesthesia Machine:
---------Ambient temperature: 10 ～ 40 ℃ ;
---------Relative humidity: no higher than 80 %;
--------Atmospheric pressure: 860 hPa ～ 1060 hPa.
--------Power requirement: ～220－240V, 50/60Hz, 40VA, to be well grounded.
--------Air source requirement: medical oxygen and laughing gas with a rated pressure ranging from 0.3 to 0.5MPa.
2. Attention: the AC power supply used for the anesthesia machine must be well grounded.
Attention: the anesthesia machine used must be equipped with a carbon dioxide monitor complying with ISO 9918：1993, an oxygen monitor complying with ISO 7767：1997 and an expiratory gas volume monitor complying with 126.96.36.199 of Medical Electrical Equipment Part II: Special Requirements for Safety and Basic Performance of Anesthesia System.
3.1 Operating mode: circulatory closed, semi-closed and semi-open.
3.2 Gas requirement: medical oxygen and nitrous oxide with a pressure ranging from 0.3 MPa to 0.5 MPa.
3.3 The maximum error of the pressure testing device should not exceed ± (4% of the full scale reading + 8% of the actual reading).
3.4 For oxygen and nitrous oxide, a special pressure regulator with a safety valve will be provided. The exhaust pressure of the safety valve should not be higher than 6 kPa.
3.5 Indication range of the oxygen and nitrous oxide flow meter: 0.1 L／min ～ 10 L／min.
When the flow rate ranges from 10% of the full scale value to 100%, the scale accuracy should be within ±10％ of the indicated value.
3.6 The flow meter is equipped with an oxygen-nitrous oxide proportional control device.
When the oxygen concentration in the mixed gas N2O/O2 conveyed by the anesthesia
machine is no lower than 20% (V/V) or the FiO2 is lower than 20%, the machine will emit an
3.7 When the oxygen pressure of the anesthesia machine is 0.20MPa±0.05MPa, the machine raises a low gas source pressure alarm which is a high priority alarm, and shuts off the nitrous oxide conveyed to the common gas outlet.
3.8 Oxygen Flush:25~75 L/Min;
3.9 Anesthetic gas concentration adjustment range of vaporizer: ０～5％, relative error ±20 %.
3.10 The exhaust pressure of the safety valve of the respiratory circuit is no higher than 6 kPa.
3.11 Anesthesia ventilator
3.11.1 Respiration Mode:IPPV,SIPPV,Manu
3.11.2 Respiration Frequency:4~40bpm
3.11.3 I/E Ratio:1:1.5~1:4
3.11.4 Tidal Volume:50~1500mL
3.11.6 Controlled ventilation and assisted ventilation switching time: 6s
3.11.7 Maximum safety pressure: ≤ 12.5 kPa.
3.11.8 Pressure limit range: 1～6 kPa
3.11.9 Airway pressure alarm: adjustment range of upper alarm limit: 0.3kPa ～ 6 kPa, permissible error ±0.2 kPa, or ±15% (whichever is greater), the machine should immediately raise a high level alarm when the airway pressure rises to the alarm value; the lower alarm limit ranges from 0.2 to 5 kPa and the permissible error is ±0.2 kPa or ±15% (whichever is greater). The machine should immediately raise a medium level alarm when the airway pressure drops to the alarm value and such condition lasts 4 to 15s.
3.11.10 Tidal volume alarm: the upper alarm limit ranges from 50 to 2000ml, the permissible error is ±20％, the adjustment range of the lower alarm limit is 0~1800ml, the permissible error is ±20％ and the machine should raise a medium level alarm.
3.11.11 Ventilation volume lower alarm limit: the adjustment range is 0~12 L/min , and the permissible error is ±20％. The ventilation volume upper alarm limit is fixed to be 25 L/min, the permissible error is ±20％ and the machine should raise a medium level alarm.
3.11.12 Power supply fault alarm: the machine should give audible alarm in case of power outage and the alarm should last more than 120s.
3.11.13 The silencing time of the audible alarm signal should be less than 120s. The alarm status should not be deactivated and visual alarm signal should not be intermittent.
3.11.14 The emergency battery should be a lead acid battery with a rated output voltage of DC 12V. The duration of the operation of the anesthetic respirator driven by the battery should be more than 60min.
4 Dimensions and Weight
Dimensions: 1580 mm×500 mm×850 mm
Weight: 75 kg.
5 Installation and Commissioning
Attention: installation, commissioning, inspection and use of the anesthesia machine should be completed by qualified professionals. This anesthesia machine has been commissioned by a professional when delivered for use. The user does not need to adjust it; otherwise it will cause unexpected fault or damage.
Company Advantages1. The materials employed in the production of MeCan ventilator price is up to the international standard.
With the improvement of social economy, MeCan has always kept up with the trend to enhance its strength in portable patient monitor industry. MeCan is a dominant provider which owns portable patient monitor. As the economy goes, MeCan has been the prominent portable patient monitor supplier in this industry.
Guangzhou MeCan Medical Limited has a group of senior design elite team and management personnel for portable patient monitor development. Guangzhou MeCan Medical Limited has a group of elite senior management team and dedicated staff. Guangzhou MeCan Medical Limited has a group of highly skilled engineers to provide the best solution on portable patient monitor. Guangzhou MeCan Medical Limited has a high-quality professional R&D team for portable patient monitor.
We are committed to the highest standards of excellence across all aspects of our business. We look for opportunities to exceed expectations and strive with determination to win and be the best at what we do. We have a clear business goal: to improve overall customer satisfaction. Instead of constantly expanding markets, we invest more in improving product quality and customer services to bring customers the product solutions to the utmost. We attach high importance to the common development of local communities. We are active in promoting communities' development. We will continue to take part in poor relief programs to drive local economic growth. We aim to be the leading provider in China. We have developed a detailed strategy to help us achieve this goal by standing out from the industry and providing the best services.
Application of the Product
MeCan's Trolley Color Doppler is widely used in the industry.
MeCan provides customers with unique solutions to meet their individual needs.